Neware 8 Channel Battery Cells Analyzer with Laptop and Software for Research
The test of polymer batteries and super capacitors need Neware special accessories . Cylindrical batteries need to be clamped between the upper and down clamps of the equipment . By moving the position of the upper accessory accurately according to the scales on both sides , you can test different capacities of Cylinders Batteries .
1 , Provide timely instructions , warnings , and intelligent support .
2 , Compliance CE Certified .
3 , Of course , high C - Rate charge - Quick charge is another reason for higher current testers become more popular than before .
4 , Windows - based software interface for easy operation .
5 , Uses standard SQL data - base management .
6 , Real - time monitoring windows and integrated graph / data windows can set various working modes for measuring capacity and cycle life of all types of rechargeable batteries , such as constant current discharge , constant current charge , constant voltage charge , rest , as well as voltage , current , time , capability , negative voltage slope , and other limit conditions .
7 , The user can define the curve by different types such as voltage - time curve , current - time curve , capacity - voltage curve ,
loop times - charge / discharge capacity curve , loop times - charge / discharge efficiency curve , etc . to make curves be compared both by visually and statistically .
For a 3000mAh cell battery , people want to charge it at 6000mA - 2C or even 9000mA - 3C current . And in some labs , researchers are trying 10C or even higher C - Rate charge or discharge .
|Current||Per Channel Current Range||Range1 , 5mA - 1A Range2 , 1A - 3A Range3 , 3A - 6A|
|Accuracy||0.05 percent of FS|
|Stability||0.05 percent of FS|
|Data Recording||Record Condition||Min Time - 100mS|
|Min Voltage - 10mV|
|Cycle||Loop Measure Range||65535 times|
|Max Steps Per Loop||254|
|Nested Loop||Nested loop Function , Max Support 3 Layers|
|Protection||Software Protection||Power - down Data Protection|
|Off - line Testing Function|
|Setting Protection Condition,Setting Parameter - Low Voltage limit , Upper Voltage limit , Low Current limit , Upper Current limit , Delay Time|