Mr. SAVIN Jean-FrancoisWe have been using Neware benches for testing our primary cells and batteries for several years. We are quite satisfied about the quality of this type of equipment and we find always good and very reactive feedback from their after-sales service to help us in specific case.
Prof. Lv LiWe have been using Neware battery testing system for many years. We are focused on different electrochemical tests of lithium-ion batteries. The Neware machine is easy to be used and the program is easy to set. We have conducted charge-discharge cycling for more than 5000 cycles and the machine showed very good stability, which helped us to obtain
Mr. Frank GroenVarious neware testing systems have been equipped our laboratory for years and serves very good. We use them for many kinds of tests. Easy to handle the tests and data is well organized for our reporting work. The most important thing is the measurement and control are quite accurate and reliable.
20V 20A 8 Channel Battery Cycle Tester Machine Neware Electric Tool
|Charge Voltage Range||100mV - 20V||Current Range||100mA - 20A|
|Min Current Record Condition||40mA||Min Voltage Record Condition||40mV|
|Input Power||2228W||Warranty||1 Year|
20A Battery Cycle Tester,
8 Channel Battery Cycle Tester,
Neware battery tester
Neware Electric Tool Battery Cycle Life Testing Machine 20V20A 8 Channel
1 . CT4000, based on Neware 4th generation testing system , entered into market in 2008 .
2 . It supports some testings as follow , EV battery pulse charge/discharge , DCIR - Direct Current Internal Resistance , cycle life and rate .
3 . CT4000 mainly applies for institutions , colleges and universities and EV battery manufacturers .
4 . Most of us are considering the voltage and current accuracies are the all important factors of a battery tester , but actually timing accuracy is of the same importance .
5 . Capacity or energy is calculated on time basis with the voltage and current measured , thus the timing accuracy will greatly affect the capacity calculated. To a general constant current or other constant testing regimes , this effect may not that easy to be observed .
6 . Neware battery testing system utilizes LAN as the main framework to transfer data . This provides the most flexibility and reliablity to your system deployment .
7 . The Neware battery tester models listed below here in different categories are the most popular ones chosen by our customers , so if you belong to one of the following category , then it might be easier for you to find out the battery tester that meets your requirement based on your peers choice .
8 . BTS4000 series entered into market in the year of 2008 . Now it is one of the most successful and most popular battery testing system in China and World . Thousands of battery manufacturers , battery research institutes and laboratories use it as their primary testing partner for their various testing activities .
|Voltage||Per Channel Voltage Range||100mV - 20V|
|Discharge Min Voltage||2.5V|
|Accuracy||0.05 percent of FS|
|Stability||0.05 percent of FS|
|Time||Current Response Time||20ms|
|Charge||Charge Mode||CCCharge , CVCharge , CCCVCharge , CPCharge|
|End Conditon||Voltage , Current , Relative Time , Capacity , -deltaV|
|Discharge||Discharge Mode||CCD , CPD , CRD|
|End Condition||Voltage , Current , Relative Time , Capacity|
|Pulse||charge||CCC , CPC|
|Discharge||CCD , CPD|
|Min Pulse Width||500ms|
|Automated Switch||Automated swith from charge to discharge for each pulse|
|End Condition||Voltage, Test Time|
|Cycle||Loop Measure Range||65535 times|
|Max Steps Per Loop||254|
|Nested Loop||Nested loop Function , Max Support 3 Layers|
|Voltage and Current Testing Sample||4 - wire connecting|
|Noise||Less than 85dB|
|A Means of Communication of Upper Machine||TCP / IP Protocol|