Mr. SAVIN Jean-FrancoisWe have been using Neware benches for testing our primary cells and batteries for several years. We are quite satisfied about the quality of this type of equipment and we find always good and very reactive feedback from their after-sales service to help us in specific case.
Prof. Lv LiWe have been using Neware battery testing system for many years. We are focused on different electrochemical tests of lithium-ion batteries. The Neware machine is easy to be used and the program is easy to set. We have conducted charge-discharge cycling for more than 5000 cycles and the machine showed very good stability, which helped us to obtain
Mr. Frank GroenVarious neware testing systems have been equipped our laboratory for years and serves very good. We use them for many kinds of tests. Easy to handle the tests and data is well organized for our reporting work. The most important thing is the measurement and control are quite accurate and reliable.
3 Electrodes High Rate Discharge Cell Tester , 5V20mA Lithium Battery Testing Equipment
|Voltage Range||0V - 5V||Current Range||5uA – 1mA 1mA – 10mA 10mA – 20mA|
|Current Response Time||500uS||Pulse||CC , CP|
5V20mA lithium battery testing equipment,
3 Electrodes lithium battery testing equipment,
lithium high rate discharge cell tester
Neware 5V20mA Battery Cell Charger Three Electrodes Material Cycler
1 . Neware 5V10mA / 20mA / 50mA is born for coin cell test , formerly its 5V1mA this modle . Students and researchers in universities and institutes just love this product , its the real star on their mind .
2 . The design of the uA current unit plays a vital role in the field of current accurate measurement , which is more convenient and more accurate .
3 . The lower range is able to output current low as 5uA , which can tell the subtle difference of various materials such electrodes , electrolyte , separator , etc .
4 . Generally , magnetoelectric galvanometer can achieve uA level , electromagnetic galvanometer can achieve mA level .
5 . Sensitive galvanometer is used for students to check weak current or tiny voltage in DC circuit in the lab . It is a kind of high sensitive magnetoelectric instrument , which can measure the tiny current .
|Data Recording||Record Frequency||10Hz|
|Charge||Charge Mode||CCCharge , CVCharge , CCCVCharge , CPCharge|
|End Conditon||Voltage , Current , Relative Time , Capacity , -deltaV|
|Discharge||Discharge Mode||CCD , CPD , CRD|
|End Condition||Voltage , Current , Relative Time , Capacity|
|Pulse||charge||CCC , CPC|
|Discharge||CCD , CPD|
|Min Pulse Width||500ms|
|Automated Switch||Automated swith from charge to discharge for each pulse|
|End Condition||Voltage, Test Time|
|Cycle||Loop Measure Range||65535 times|
|Max Steps Per Loop||254|
|Nested Loop||Nested loop Function , Max Support 3 Layers|
|Voltage and Current Testing Sample||4 - wire connecting|
|A Means of Communication of Upper Machine||TCP / IP Protocol|
|Communication Interface||Ethernet Port|
|Per Unit Main Channels No.||8|
|Dimensions||3U , 480 – 404 - 130mm|