Mr. SAVIN Jean-FrancoisWe have been using Neware benches for testing our primary cells and batteries for several years. We are quite satisfied about the quality of this type of equipment and we find always good and very reactive feedback from their after-sales service to help us in specific case.
Prof. Lv LiWe have been using Neware battery testing system for many years. We are focused on different electrochemical tests of lithium-ion batteries. The Neware machine is easy to be used and the program is easy to set. We have conducted charge-discharge cycling for more than 5000 cycles and the machine showed very good stability, which helped us to obtain
Mr. Frank GroenVarious neware testing systems have been equipped our laboratory for years and serves very good. We use them for many kinds of tests. Easy to handle the tests and data is well organized for our reporting work. The most important thing is the measurement and control are quite accurate and reliable.
Neware 18650 Battery Charger Capacity Tester 10Hz Frequency
|End Conditon||Voltage , Current , Relative Time , Capacity||Charge||CCC , CVC , CCCCVC , CPC|
|Stability||0.05percent Of FS||Hardware Protection||Reverse Protection Model|
|Software Protection||Power - Down Data Protection||Frequency||10Hz|
10Hz 18650 Battery Charger Capacity Tester,
Neware 18650 Battery Charger Capacity Tester,
Neware battery charge discharge tester
Neware 18650 Battery Rate Charger Capacity and Cycle Life Tester
1 . Neware battery testing system utilizes LAN as the main framework to transfer data. This provides the most flexibility and reliablity to your system deployment.
2 . When you look down into Neware batter, its unique Middle Machine architecture provides another enhancement layer to the data security issue. Moreover, this MM is an embedded system that allows you run tests even your LAN is down.
3 . As we go deeper, dual embedded systems will have large redundancy to handle controls and data transfer.
4 . HPPC , Capacity , Cycle Life , Formation and Grading , Over - charge , Over - discharge
5 . Most of us are considering the voltage and current accuracies are the all important factors of a battery tester , but actually timing accuracy is of the same importance .
6 . Capacity or energy is calculated on time basis with the voltage and current measured , thus the timing accuracy will greatly affect the capacity calculated . To a general constant current or other constant testing regimes , this effect may not that easy to be observed .
7 . Left is from Neware , right is from another supplier in the marke .
8 . Market - Proven Testing System
9 . Less than 20ms Response Time to Meet Rapidly Changing Power Requirements .
10 . 0.1 percent Full - scale Accuracy for Current and Voltage .
11 . 10Hz Data Acquisition Frequency , 100Hz Optional .
12 . 500ms Minimum Pulse Width More Than 1 Million Records Can Be Processed .
|Input Resistance||More than 1MΩ|
|Channel Features||Constant Current Source and Constant Voltage Source with Independent pairs of Closed - loop Structure|
|Channel Control Mode||Independent Control|
|Voltage||Per Channel Voltage Range||25mV - 5V|
|Discharge Min Voltage||2.5V|
|Accuracy||0.05percent of FS|
|Current||Per Channel Current Range||0.1A - 20A|
|Accuracy||0.05percent of FS|
|Power||Per Channel Output Power||100W|
|Data Recording||Record Condition||Time delta T , 100mS|
Constant Current Charge , Constant Voltage Charge ,
Constant Current and Constant Voltage Charge , CPC
|End Conditon||Voltage , Current , Relative Time , Capacity|
|Discharge||Discharge Mode||CCD , CPD , CRD|
|End Condition||Voltage , Current , Relative Time , Capacity|
|Automated Switch||Automated swith from charge to discharge for each pulse|
|End Condition||Voltage , Test Time|
|Cycle||Loop Measure Range||65535 times|
|Max Steps Per Loop||254|
|Nested Loop||Nested loop Function , Max Support 3 Layers|
|Protection||Software Protection||Power - down Data Protection|
|Off - line Testing Function|
|Setting Protection Condition , Setting Parameter - Low Voltage limit / Upper Voltage limit / Low Current limit / Upper Current limit / Delay Time|
|Hardware Protection||Additional Anti - reverse Protection Model|