Battery Cycle Tester
Battery Cell Cycler
Battery Cycler System
Lithium Ion Battery Analyzer
Battery Capacity Discharge Tester
18650 Battery Charger Capacity Tester
8 Channel Battery Analyzer
Battery Cell Tester
Regenerative Battery Pack Test System
Neware Battery Test System
EV Battery Testing Equipment
Battery Grading System
Battery Test Chamber
Mr. SAVIN Jean-FrancoisWe have been using Neware benches for testing our primary cells and batteries for several years. We are quite satisfied about the quality of this type of equipment and we find always good and very reactive feedback from their after-sales service to help us in specific case.
Prof. Lv LiWe have been using Neware battery testing system for many years. We are focused on different electrochemical tests of lithium-ion batteries. The Neware machine is easy to be used and the program is easy to set. We have conducted charge-discharge cycling for more than 5000 cycles and the machine showed very good stability, which helped us to obtain
Mr. Frank GroenVarious neware testing systems have been equipped our laboratory for years and serves very good. We use them for many kinds of tests. Easy to handle the tests and data is well organized for our reporting work. The most important thing is the measurement and control are quite accurate and reliable.
LIR1254 Battery Cycler System
|Discharge Voltage||-5V - 5V||Record Condition||Range 1 , 2uA Range 2 , 0.02mA Range 3 , 0.04mA|
|Record Frequency||10Hz||Nested Loop||Nested Loop Function , Max Support 3 Layers|
|Main Channels No.||8||Max Steps Per Loop||254|
LIR1254 Battery Cycler System,
TWS Battery Cycler System
Neware TWS LIR1254 Cell Battery Cycler System Rechargeable Battery Tester
1 . Sensitive galvanometer is used for students to check weak current or tiny voltage in DC circuit in the lab . It is a kind of high sensitive magnetoelectric instrument , which can measure the tiny current .
2 . Testing with the uA current unit is more accurate and more convenient .
3 . The lower range is able to output current low as 5uA , which can tell the subtle difference of various materials such electrodes , electrolyte , separator , etc .
4 . Generally , magnetoelectric galvanometer can achieve uA level , electromagnetic galvanometer can achieve mA level .
5 . dQ / dV differential capacity curve , the peak on the curve indicates that there is a voltage plateau on the charge and discharge curve , and different peaks represent different electrochemical reactions .
6 . The design of the uA current unit plays a vital role in the field of current accurate measurement , which is more convenient and more accurate .
7 . Neware 5V10mA / 20mA / 50mA is born for coin cell test , formerly its 5V1mA this modle . Students and researchers in universities and institutes just love this product , its the real star on their mind .
|Charge||Charge Mode||CCCharge , CVCharge , CCCVCharge , CPCharge|
|End Conditon||Voltage , Current , Relative Time , Capacity , -deltaV|
|Discharge||Discharge Mode||CCD , CPD , CRD|
|End Condition||Voltage , Current , Relative Time , Capacity|
|Pulse||charge||CCC , CPC|
|Discharge||CCD , CPD|
|Min Pulse Width||500ms|
|Automated Switch||Automated swith from charge to discharge for each pulse|
|End Condition||Voltage, Test Time|
|Cycle||Loop Measure Range||65535 times|
|Max Steps Per Loop||254|
|Nested Loop||Nested loop Function , Max Support 3 Layers|
|Protection||Software Protection||Power - down Data Protection|
|Off - line Testing Function|
|Setting Protection Condition,Setting Parameter - Low Voltage limit , Upper Voltage limit , Low Current limit , Upper Current limit , Delay Time|
|IP protection level||Protection level IP20|
|Voltage and Current Testing Sample||4 - wire connecting|
|Per Unit Main Channels No.||8|