Mr. SAVIN Jean-FrancoisWe have been using Neware benches for testing our primary cells and batteries for several years. We are quite satisfied about the quality of this type of equipment and we find always good and very reactive feedback from their after-sales service to help us in specific case.
Prof. Lv LiWe have been using Neware battery testing system for many years. We are focused on different electrochemical tests of lithium-ion batteries. The Neware machine is easy to be used and the program is easy to set. We have conducted charge-discharge cycling for more than 5000 cycles and the machine showed very good stability, which helped us to obtain
Mr. Frank GroenVarious neware testing systems have been equipped our laboratory for years and serves very good. We use them for many kinds of tests. Easy to handle the tests and data is well organized for our reporting work. The most important thing is the measurement and control are quite accurate and reliable.
Neware BTS9000 Lithium Ion Battery Analyzer Coin Cells Pouch Tester
Contact me for free samples and coupons.
Wechat: 0086 18588475571
If you have any concern, we provide 24-hour online help.x
|Database||MySQL||Communication Interface||Ethernet Port|
|Per Unit Main Channels No.||4||Nested Loop||Nested Loop Function , Max Support 3 Layers|
|Current Response Time||100μS Below||Current Delta I||100nA|
Neware Lithium Ion Battery Analyzer,
BTS9000 Lithium Ion Battery Analyzer,
Neware lithium ion battery tester
Neware BTS9000 Lithium Ion Battery Cycler Coin Cells Pouch Analyzer
CT - 9004 Cutting - edge Technology for Battery Material Research
1 . CT - 9004 , as our latest battery testing system , brings you up - to- date , sophisticated hardware technology and brand - new software experience .
2 . Its characterized with a maximum data acquisition frequency of 1000Hz , 4 automated - switch output ranges , 0.02percent accuracy and a minimum pulse width of 400μs .
GSM TEST / PULSE TEST / DCIR TEST
3 . BTS9000 is based on Neware sixth - generation testing system . It's designed for battery material research , high - precision tests, pulse charge/discharge tests , DCIR test , and cycle life test .
4 . As the most cutting - edge equipment for battery material research , BTS9000 comes with extremely high accuracy 0.02 percent of FS , amazing hardware response speed - less than 100μs , wide range of applications , high - speed testing data acquisition and many other industry - proven characteristics , its your ultimate choice for battery material research .
5 . We designed and shipped our sophisticated series which is characterized with a max . data acquisition frequency of 1000Hz , a minimum pulse width reaching 400us , an accuracy of 0.02percent of FS and auto multi - current ranges enabled .
43,000 Hours - 5 years
6 . Uninterrupted operation hours of 43000 hours , and equipment still outputs and measures accurately .
Neware CT - 9000 is the best ever battery tester from Neware , it came into the market with extremely high accuracy , Amazingly quick hardware response time , high - speed data acquisition frequency and many other industry - proven characteristics , it can tell the subtle difference of all kinds of battery materials , it helps you to find the future battery material and solution . CT - 9000 is mainly designed for Battery research insitutues , universities and laboratories of battery companies.
|Project index||Parameter index|
|Voltage||Per Channel Voltage Range||0 V~5V|
|Min discharge voltage||0.7 V - 0.5m cable , 5A discharge|
|Accuracy||0.02percent of FS|
|Stability||0.005percent of FS|
|Current||Current range||range1 , 0.1uA -150uA|
|range2 , 140uA - 5mA|
|range3 , 4.5mA - 150mA|
|range4 , 140mA - 5A|
|Accuracy||0.02 percent of FS|
|range1 , 30nA|
|range2 , 1uA|
|range3 , 30uA|
|range4 , 1mA|
|Stability||0.005 percent of FS|
|Pulse mode||Charge||CC , CP mode|
|Discharge||CC , CP mode|
|Min pulse width||400μs|
|Pulse number||Single pulse working step supports 16 different pulse segments|
|End Condition||Voltage , Relative Time|
|DCIR||Support for the calculation of the point of the DCIR|
|A Means of Communication of Upper Machine||TCP / IP Protocol , 100M Ethernet|
|Data Output||EXCEL2003 , 2010 , TXT|
|Communication Interface||Ethernet Port|
|Per Unit Main Channels No.||4|
|Dimensions||WDH , 585 - 575 - 130mm|