Mr. SAVIN Jean-FrancoisWe have been using Neware benches for testing our primary cells and batteries for several years. We are quite satisfied about the quality of this type of equipment and we find always good and very reactive feedback from their after-sales service to help us in specific case.
Prof. Lv LiWe have been using Neware battery testing system for many years. We are focused on different electrochemical tests of lithium-ion batteries. The Neware machine is easy to be used and the program is easy to set. We have conducted charge-discharge cycling for more than 5000 cycles and the machine showed very good stability, which helped us to obtain
Mr. Frank GroenVarious neware testing systems have been equipped our laboratory for years and serves very good. We use them for many kinds of tests. Easy to handle the tests and data is well organized for our reporting work. The most important thing is the measurement and control are quite accurate and reliable.
Intelligent Lock Battery Cycler System , 10V10A Battery Discharge Test Equipment
|Charge Voltage Range||50mV - 10V||Current Range||50mA - 10A|
|Min Current Record Condition||20mA||Min Voltage Record Condition||20mV|
|Input Power||1215W||Channel Number||64 Per Rack|
Intelligent Lock battery discharge test equipment,
10V battery discharge test equipment,
10A battery cycler system
Neware 10V10A Battery Capacity Discharge Tester Intelligent Lock Battery Cycler
1 , It supports some testings as follow , EV battery pulse charge/discharge , DCIR - Direct Current Internal Resistance , cycle life and rate .
2 , Battery researchers and scientists, battery manufactures , importers and exporters, you can always choose the most appropriate Neware battery testers for battery quality check , battery performance tests, battery material research , and battery mass production lines .
3 . Most of us are considering the voltage and current accuracies are the all important factors of a battery tester , but actually timing accuracy is of the same importance .
4 . Capacity or energy is calculated on time basis with the voltage and current measured , thus the timing accuracy will greatly affect the capacity calculated. To a general constant current or other constant testing regimes , this effect may not that easy to be observed .
5 . Neware battery testing system utilizes LAN as the main framework to transfer data . This provides the most flexibility and reliablity to your system deployment .
6 . When you look down into Neware battery tester , its unique Middle Machine architecture provides another enhancement layer to the data security issue . Moreover , this MM is an embedded system that allows you run tests even your LAN is down .
|Channel Features||Constant Current Source and Constant Voltage Source with Independent pairs of Closed - loop Structure|
|Channel Control Mode||Independent Control|
|Time||Current Response Time||20ms|
|Charge||Charge Mode||CCCharge , CVCharge , CCCVCharge , CPCharge|
|End Conditon||Voltage , Current , Relative Time , Capacity , -deltaV|
|Discharge||Discharge Mode||CCD , CPD , CRD|
|End Condition||Voltage , Current , Relative Time , Capacity|
|Pulse||charge||CCC , CPC|
|Discharge||CCD , CPD|
|Min Pulse Width||500ms|
|Automated Switch||Automated swith from charge to discharge for each pulse|
|End Condition||Voltage, Test Time|
|Cycle||Loop Measure Range||65535 times|
|Max Steps Per Loop||254|
|Nested Loop||Nested loop Function , Max Support 3 Layers|
|Protection||Software Protection||Power - down Data Protection|
|Off - line Testing Function|
|Setting Protection Condition,Setting Parameter - Low Voltage limit , Upper Voltage limit , Low Current limit , Upper Current limit , Delay Time|
|IP protection level||Protection level IP20|
|Data Output||EXCEL , TXT , Graph|
|Communication Interface||Ethernet Port|
|Per Unit Main Channels No.||8|