Mr. SAVIN Jean-FrancoisWe have been using Neware benches for testing our primary cells and batteries for several years. We are quite satisfied about the quality of this type of equipment and we find always good and very reactive feedback from their after-sales service to help us in specific case.
Prof. Lv LiWe have been using Neware battery testing system for many years. We are focused on different electrochemical tests of lithium-ion batteries. The Neware machine is easy to be used and the program is easy to set. We have conducted charge-discharge cycling for more than 5000 cycles and the machine showed very good stability, which helped us to obtain
Mr. Frank GroenVarious neware testing systems have been equipped our laboratory for years and serves very good. We use them for many kinds of tests. Easy to handle the tests and data is well organized for our reporting work. The most important thing is the measurement and control are quite accurate and reliable.
LFP 16CH Bts Neware Battery Test System 10C Rate Discharge
|Voltage Range||25mV - 5V||Current Range||400mA - 80A|
|Input Power||11342W||Min Voltage Resolution||10mV|
|Min Current Resolution||160mA||Per Unit Channel Number||16|
16CH Neware Battery Test System,
10C Rate Neware Battery Test System,
LFP high rate discharge cell tester
16CH Neware Battery Testing System LFP 10C Rate Discharge Equipment
DCIR - Direct Current Internal Resistance
The variation of DCIR has a great influence on battery discharge performance , especially for high power batteries . In general , the better the battery , the lower the internal resistance . Therefore , most battery manufacturers identify DCIR as a key indicator to evaluate battery quality . The method of DCIR measurement is to inject high current into the anodeand cathode of the battery in a short period , and then record the changes of battery voltage and charge / discharge current . To calculate DCIR by Ohm Law , voltage variation is divided by current variation .
Direct current internal resistance of batteries is the r esistance of current flowing through the battery , with the standard unit of measurement called the ohm .The value of DCIR is not fixed , and varies depending on multiple factors , such as battery materials , electrolyte concentration , temperature , and depth of discharge .
Neware battery testers provide good accuracies both for voltage and current . This ensures you get the most reliable data from your tests .
A measurement system can be accurate but not precise , precise but not accurate , neither , or both .
Neware battery testers provide good prrecision meanwhile . To get an overall performance of a measurement system , Measurement system analysis MSA is usually introduced for the study .
1 , Windows-based software interface for easy operation .
2 , Uses standard SQL data - base management .
3 , Real - time monitoring windows and integrated graph / data windows can set various working modes for measuring capacity and cycle life of all types of rechargeable batteries , such as:constant current discharge , constant current charge , constant voltage charge , rest , as well as voltage , current , time , capability , negative voltage slope , and other limit conditions .
4 , The user can define the curve by different types such as voltage - time curve , current - time curve , capacity - voltage curve ,
loop times - charge / discharge capacity curve , loop times - charge / discharge efficiency curve , etc . to make curves be compared both by visually and statistically .
5 , Provide timely instructions , warnings, and intelligent support .
6 , Compliance CE Certified .
|resolution||AD 16bit , DA 16bit|
|Input Resistance||More than 1MΩ|
|Channel Features||Constant Current Source and Constant Voltage Source with Independent pairs of Closed - loop Structure|
|Channel Control Mode||Independent Control|
|Voltage||Per Channel Voltage Range||25mV - 5V|
|Discharge Min Voltage||2.5V|
|Accuracy||0.1 percent of FS|
|Stability||0.1 percent of FS|
|Time||Current Response Time||20ms|
|Charge||Charge Mode||CCCharge , CVCharge , CCCVCharge , CPCharge|
|End Conditon||Voltage , Current , Relative Time , Capacity , -deltaV|
|Discharge||Discharge Mode||CCD , CPD , CRD|
|End Condition||Voltage , Current , Relative Time , Capacity|
|Pulse||charge||CCC , CPC|
|Discharge||CCD , CPD|
|Min Pulse Width||500ms|
|Automated Switch||Automated swith from charge to discharge for each pulse|
|End Condition||Voltage, Test Time|
|Cycle||Loop Measure Range||65535 times|
|Max Steps Per Loop||254|
|Nested Loop||Nested loop Function , Max Support 3 Layers|
|Protection||Software Protection||Power - down Data Protection|
|Off - line Testing Function|
|Setting Protection Condition,Setting Parameter - Low Voltage limit , Upper Voltage limit , Low Current limit , Upper Current limit , Delay Time|
|IP protection level||Protection level IP20|
|Voltage and Current Testing Sample||4 - wire connecting|
|A Means of Communication of Upper Machine||TCP / IP Protocol|
|Data Output||EXCEL , TXT , Graph|
|Communication Interface||Ethernet Port|
|Per Unit Main Channels No.||16|