Mr. SAVIN Jean-FrancoisWe have been using Neware benches for testing our primary cells and batteries for several years. We are quite satisfied about the quality of this type of equipment and we find always good and very reactive feedback from their after-sales service to help us in specific case.
Prof. Lv LiWe have been using Neware battery testing system for many years. We are focused on different electrochemical tests of lithium-ion batteries. The Neware machine is easy to be used and the program is easy to set. We have conducted charge-discharge cycling for more than 5000 cycles and the machine showed very good stability, which helped us to obtain
Mr. Frank GroenVarious neware testing systems have been equipped our laboratory for years and serves very good. We use them for many kinds of tests. Easy to handle the tests and data is well organized for our reporting work. The most important thing is the measurement and control are quite accurate and reliable.
18650 Lithium Ion Battery Charger Tester Analyzer 5V12A Power 8 Channel
|Per Channel Current Range||1 . 0.5mA - 0.1A 2 . 0.1A - 3A 3 , 3A - 6A||Stability||0.05 Percent Of FS|
|Dimensions||480 - 404 - 130mm||Communication||Ethernet Port|
|IP Protection||Protection Level IP20||Noise||85dB|
18650 battery charger tester analyzer,
5V12A battery charger tester analyzer,
8 Channel battery charger tester analyzer
Neware 5V12A 8 Channel Battery Testing System For 18650 Lithium Ion Battery
1 . Nowadays battery capacity density becomes higher and higher , thats the trend of this whole energy industry , people need smaller dimension battery to contain more energy . New technology and upgraded producing process make this idea come into reality . Of course , high C - Rate charge - Quick charge is another reason for higher current testers become more popular than before . For a 3000mAh cell battery , people want to charge it at 6000mA - 2C or even 9000mA - 3C current . And in some labs , researchers are trying 10C or even higher C - Rate charge or discharge . In this situation , 5V12A is the very rising star for both battery manufactures and researchers .
2 . To meet the various requirements adopted in advanced battery research , our equipments can automatically switch the current range in accordance with user settings to maintain consistent accuracy , and can be fitted with additional hardware to provide a number of optional features and capabilities .
3 . The model 5V6A can be configured with multi - current range channels , which have a current control range of 0.5mA to 0.1A , 0.1A to 3A , 3A to 6A as standard . The accuracy for current and voltage on these channels is 0.05 percent of FS . From the base system that provides the highest level of accuracy and time resolution , features can be added to allow the 5V6A test system to perform virtualy many type of test , with many type of products – cylindrical , pouch and coin cells , super – capacitors , etc , and chemistries .
4 . Neware 5V6A is the most popular battery tester ever , more than 50 percent customers would choose this model as their main products . For customers who came across Neware the first time , 5V6A is also strongly suggested as their tryout demo machine . Neware 5V6A is powerful , reliable , it works on the very popular coin cell , cylindrical cell and prismatic cells , etc . It can be used for battery formation and grading , cycle - life test , performance and so on . From year 2016 , dual range 5V6A was successfully replaced the traditional 5V6A model . Higher range 100mA - 3A - 6A , makes it the best choice for cycle - life test and C - Rate test on 3C battery cells . Lower range 0.5mA - 100mA - 1A , makes 5V6A great for small pouch , coin cell and small battery test .
|resolution||AD 16bit , DA 16bit|
|Input Resistance||More than 1MΩ|
|Channel Features||Constant Current Source and Constant Voltage Source with Independent pairs of Closed - loop Structure|
|Channel Control Mode||Independent Control|
|Voltage||Per Channel Voltage Range||1.5V - 5V|
|Discharge Min Voltage||1.5V|
|Accuracy||0.05 percent of FS|
|Stability||0.05 percent of FS|
|Power||Per Channel Output Power||30W|
|Stop current||Range 1 , 2mA Range 2 , 12mA Range 3 , 24mA|
|Stability||0.1percent of FS|
|Time||Current Response Time||1ms|
|Data Recording||Record Condition||Min Time - 100mS|
|Min Voltage - 10mV|
|Min Current - Range 1 , 2mA Range 2 , 12mA Range 3 , 24mA|
|Charge||Charge Mode||CCCharge , CVCharge , CCCVCharge , CPCharge|
|End Conditon||Voltage , Current , Relative Time , Capacity , -deltaV|
|Discharge||Discharge Mode||CCD , CPD , CRD|
|End Condition||Voltage , Current , Relative Time , Capacity|
|Pulse||charge||CCC , CPC|
|Discharge||CCD , CPD|
|Min Pulse Width||500ms|
|Automated Switch||Automated swith from charge to discharge for each pulse|
|End Condition||Voltage, Test Time|
|Protection||Software Protection||Power - down Data Protection|
|Off - line Testing Function|
|Setting Protection Condition,Setting Parameter - Low Voltage limit , Upper Voltage limit , Low Current limit , Upper Current limit , Delay Time|
|IP protection level||Protection level IP20|
|A Means of Communication of Upper Machine||TCP / IP Protocol|
|Data Output||EXCEL , TXT , Graph|
|Communication Interface||Ethernet Port|
|Per Unit Main Channels No.||8|
|Dimensions||3U , 480 – 330 - 180mm|