Neware 25L Mini Constant Tempearture Chamber For Battery Test
Neware environmental test chamber - also knows as high and low temperature box , high temperature box , incubator and explosion proof box - is researched and produced by Neware for battery testing in new energy industry , which can provide the required high temperature , low temperature and constant temperature environment for battery charge , discharge and safety performance testing , and it can simulate climate environment for researchers to do tests in lab . Neware environmental test chamber is also a reliable equipment for all companies such as electronics , electricians , instruments , materials and semiconductor , as well as for departments such as schools , medicine , chemical , environmental protection , agriculture , livestock and aquatic products , etc .
Mini Constant Tempearture Chamber
Neware Constant Temperature Test Chamber aims to test battery performance under constant temperature by simulating room temperature condition . it can analyze and evaluate whether the battery can meet required international standard and provides a good reference in product research and development stage . It’s widely applied on Electronics , Telecommunication , Automobile , Packaging , Aerospace , Electric products , Pharmaceuticals industry , etc .
|Thermal insulation enclosure||
Outer wall material , high quality cold rolled steel plate, surface spray and paint treatment .
Inner wall material , stainless steel SUS304
Box insulation material , rigid polyurethane foam plus glass wool - insulation thickness 100mm .
Door insulation material , glass wool
|Air conditioning channel||
Centrifugal fan , flameproof type
Evaporator and dehumidifier , etc .
|Standard configuration of test chamber||
Lead hole , can be increased , 4 holes with diameter of 100 mm
With soft rubber plug , 2 on the left and right sides of the box
Casters , 4 with adjusting feet
Observation window: multi layer hollow electric heating film heating anti fog observation window - located on the door
The electric mist eliminator inside the glass can provide the best observation line of sight .
One Lighting lamp
Single hinged door - left hinge, right handle , equipped with observation window , lighting lamp
Window frame / door frame anti condensation electric heating device, double silicone rubber sealing strip
|control panel||Controller display screen , over temperature protection setter , etc|
|Refrigeration unit room||Refrigeration unit , water tray , drain hole and condensing fan|
|Power distribution control cabinet||
One main power leakage circuit breaker, distribution board , exhaust fan and Ethernet physical interface
Temperature and humidity controller , AC contactor , circuit breaker , thermal relay
Temperature limiting protector , solid state relay and transformer , etc
|heater||Fin heater , heater control mode - contactless equal period pulse width modulation , SSR solid state relay|
|Power line hole and drain hole||Located on the back of the box|
|Explosion proof pressure relief port , if have||It is located at the top of the box and automatically opens when the pressure of the test space exceeds the set pressure|
|Controller model||Professional temperature and humidity controller|
|monitor||High definition color LCD touch screen|
|Operation mode||Program mode and setting mode|
|Setting mode||Color touch, human-computer interaction, Chinese / English interface|
Anti integral saturation PID
BTC balance temperature control mode
|Temperature measurement mode||Class A armored PT100 sensor|
|Display accuracy||Temperature - 0.01 Celsius degree , Time - 1 min|
|Over temperature protection||Independent over temperature protector , when the temperature of the working room exceeds the temperature set by the protection device , it will stop and send alarm signal|
|Interconnection with battery testing equipment|
|Device hardware connection||BTS computer software , battery detection equipment and test box are interconnected by channel line and data communication line|
|others||During the test , opening the door of the test chamber will cause the temperature fluctuation in the chamber , During the test , if the door is opened for many times or for a long time , or the test sample emits wet steam , the heat exchanger of the refrigeration system may freeze and fail to work normally|