Mr. SAVIN Jean-FrancoisWe have been using Neware benches for testing our primary cells and batteries for several years. We are quite satisfied about the quality of this type of equipment and we find always good and very reactive feedback from their after-sales service to help us in specific case.
Prof. Lv LiWe have been using Neware battery testing system for many years. We are focused on different electrochemical tests of lithium-ion batteries. The Neware machine is easy to be used and the program is easy to set. We have conducted charge-discharge cycling for more than 5000 cycles and the machine showed very good stability, which helped us to obtain
Mr. Frank GroenVarious neware testing systems have been equipped our laboratory for years and serves very good. We use them for many kinds of tests. Easy to handle the tests and data is well organized for our reporting work. The most important thing is the measurement and control are quite accurate and reliable.
Separate Type 200L 160CH Coin Cell Battery Test Chamber
|Inner Box||547mm - 637mm - 847mm||Inner Material||Stainless Steel SUS304|
|Application||Constant Temperature Test Of Button Cell||Control Panel||Touch Control Buttons|
|Cooling Mode||Air Cooled||Setting Method||Touch Key|
Coin Cell Battery Test Chamber,
160CH Battery Test Chamber,
200L Battery Cell Testing Equipment
60Celsius Degree Neware Separate Type 200L 160CH Coin Cell Battery Test Chamber
Constant Temperature Chamber
1 . Neware environmental test chamber aims to test battery performance under temperature by simulating harsh climate conditions .
2 . It can analyze and evaluate whether the battery can meet the required international standard and provides a good reference in the product R&D stage .
3 . ITs widely applied to Electronics , Telecommunication , Automobile , Packaging , Aerospace , Electric products , Pharmaceuticals industry , etc .
Neware Thermal Chamber Highlights:
Commuicated with our Neware software via TCP/IP port, and connected with Neware tester;
A varity of dimension range could be made;
Widely temperature range could be supported;
Support various different batteries to test;
Easy to operate the chamber;
Standard protection included when gassing or over thermal issue.
Buid-in over temperature protection and software protection limite to ensure your test safty.
|Standard configuration of test chamber||
Lead hole , can be increased , 4 holes with diameter of 100 mm
With soft rubber plug , 2 on the left and right sides of the box
Casters , 4 with adjusting feet
Observation window: multi layer hollow electric heating film heating anti fog observation window - located on the door
The electric mist eliminator inside the glass can provide the best observation line of sight .
One Lighting lamp
Single hinged door - left hinge, right handle , equipped with observation window , lighting lamp
Window frame / door frame anti condensation electric heating device, double silicone rubber sealing strip
|heater||Fin heater , heater control mode - contactless equal period pulse width modulation , SSR solid state relay|
|Power line hole and drain hole||Located on the back of the box|
|Controller model||Professional temperature and humidity controller|
|monitor||High definition color LCD touch screen|
|Operation mode||Program mode and setting mode|
|Setting mode||Color touch, human-computer interaction, Chinese / English interface|
Anti integral saturation PID
BTC balance temperature control mode
|Temperature measurement mode||Class A armored PT100 sensor|
|Display accuracy||Temperature - 0.01 Celsius degree , Time - 1 min|
|Over temperature protection||Independent over temperature protector , when the temperature of the working room exceeds the temperature set by the protection device , it will stop and send alarm signal|
|Interconnection with battery testing equipment|
|Device hardware connection||BTS computer software , battery detection equipment and test box are interconnected by channel line and data communication line|
|others||During the test , opening the door of the test chamber will cause the temperature fluctuation in the chamber , During the test , if the door is opened for many times or for a long time , or the test sample emits wet steam , the heat exchanger of the refrigeration system may freeze and fail to work normally|