Battery Cycle Tester
Battery Cell Cycler
Battery Cycler System
Lithium Ion Battery Analyzer
Battery Capacity Discharge Tester
18650 Battery Charger Capacity Tester
8 Channel Battery Analyzer
Battery Cell Tester
Regenerative Battery Pack Test System
Neware Battery Test System
EV Battery Testing Equipment
Battery Grading System
Battery Test Chamber
Mr. SAVIN Jean-FrancoisWe have been using Neware benches for testing our primary cells and batteries for several years. We are quite satisfied about the quality of this type of equipment and we find always good and very reactive feedback from their after-sales service to help us in specific case.
Prof. Lv LiWe have been using Neware battery testing system for many years. We are focused on different electrochemical tests of lithium-ion batteries. The Neware machine is easy to be used and the program is easy to set. We have conducted charge-discharge cycling for more than 5000 cycles and the machine showed very good stability, which helped us to obtain
Mr. Frank GroenVarious neware testing systems have been equipped our laboratory for years and serves very good. We use them for many kinds of tests. Easy to handle the tests and data is well organized for our reporting work. The most important thing is the measurement and control are quite accurate and reliable.
BR2450 Battery Pack Test Equipment , Neware Lifepo4 Capacity Tester
|Discharge Voltage||-5V - 5V||Current Cover||5uA – 1mA 1mA – 25mA 25mA – 50mA|
|UNIT SIZE||480mm – 404 Mm- 130mm||End Conditon||-deltaV|
|Channel Number||8||Max Steps Per Loop||254|
BR2450 battery pack test equipment,
Neware battery pack test equipment,
Neware lifepo4 capacity tester
Neware BR2450 Battery Cycle life Test System Coin Cells Capacity Cycler
A . The design of the uA current unit plays a vital role in the field of current accurate measurement , which is more convenient and more accurate . Testing with the uA current unit is more accurate and more convenient .
B . The lower range is able to output current low as 5uA , which can tell the subtle difference of various materials such electrodes , electrolyte , separator , etc .
C . Generally , magnetoelectric galvanometer can achieve uA level , electromagnetic galvanometer can achieve mA level .
D . Sensitive galvanometer is used for students to check weak current or tiny voltage in DC circuit in the lab . It is a kind of high sensitive magnetoelectric instrument , which can measure the tiny current .
|resolution||AD 16bit , DA 16bit|
|Voltage||Per Channel Voltage Range||0V - 5V|
|Discharge Min Voltage||0V|
|Current||Per Channel Current Range||Range1 , 5uA – 1mA Range2 , 1mA – 10mA Range3 , 10mA – 20mA|
|Power||Per Channel Output Power||0.25W|
|Stop current||Range 1 , 2uA Range 2 , 0.02mA Range 3 , 0.04mA|
|Data Recording||Record Condition||Min Time - 100mS|
|Min Voltage - 10mV|
|Min Current - Range 1 , 2uA Range 2 , 0.02mA Range 3 , 0.04mA|
|Charge||Charge Mode||CCCharge , CVCharge , CCCVCharge , CPCharge|
|End Conditon||Voltage , Current , Relative Time , Capacity , -deltaV|
|Discharge||Discharge Mode||CCD , CPD , CRD|
|End Condition||Voltage , Current , Relative Time , Capacity|
|Pulse||End Condition||Voltage, Test Time|
|Cycle||Loop Measure Range||65535 times|
|Max Steps Per Loop||254|
|Nested Loop||Nested loop Function , Max Support 3 Layers|
|Voltage and Current Testing Sample||4 - wire connecting|
|Per Unit Main Channels No.||8|
|Dimensions||3U , 480 – 404 - 130mm|